Tags64-pixels adapter AVR battery bulb car customs DSO fireworks firmware good old days hack happy new year hardware IKEA import lecroy LED light lighting linux MBI5168 modding open source oscilloscope PCB power supply prototype PWM rant remote review RGB RoHS Samtid Seeedstudio SMD soldering teardown trigger upgrade WaveAce 224 weller WS2812B youyue
Tag Archives: WaveAce 224
Thanks again to everybody who was involved: eevblog and LeCroy’s current CEO Mr. Reslewic. A special thank-you goes to Hilary Lustig (also with LeCroy) for getting the machinery started. I’m glad LeCroy have handled it that way. This finally concludes … Continue reading
Today I received a call by DHL, which confirmed the pickup day and time. The WaveAce 224 was successfully picked up today at 12:09 and is now on its way to LeCroy Switzerland.
The sender was the current CEO of LeCroy, Thomas H. Reslewic. I’m glad they have finally taken action. I was offered a full refund. I hope they will also look into all the WaveAce 224 issues I’ve documented so far … Continue reading
This is a ‘close relative’ to the ‘eternal boot screen’. It is also caused by the all-mighty triggering system, which governs the whole oscilloscope – even when it shouldn’t. One of the most annoying things is that this can block … Continue reading
Usually when I turn it on it shows the blue Lecroy splash screen, thinks for a while, clicks a few relays, beeps, and then show the usual screen, channel information, grid and so forth. It got a bit warm today.
Will it ever end? How many bugs are there still to be discovered… A word of warning: there will be some strong language in the post. If anybody affiliated with Lecroy should read this, please feel offended – not as … Continue reading
Another quirk. I seem to have a talent to upset this device. Showing a signal that isn’t there should not happen though.
Automatic cursors are supposed to show what and where stuff is measured on a waveform when automatic measurements are used. This is pretty useful, if it works as expected. Suffice to say that it doesn’t always do that – unfortunately.
Got the ‘pulse-trigger’ to trigger. This time I don’t care if the set pulse-length is actually accurate, I just wanted it to trigger on the ‘thing’ and it did. Fine – happy happy joy joy! Then I felt like moving … Continue reading
I managed to damage the alligator clip on one of my probe’s ground cables. It was probably a bad one and it fatigued. I replaced the alligator clip with one I found in a box. During this procedure it turned … Continue reading
Where art thou? This time I have discovered issues that are trigger-related, but more suitably filed under “general displaying issues”. Still using the “1.01.01.18R2” firmware. Just have a look at the video. No more words…
Today I got informed that the pulse-trigger ‘bug’ has been confirmed and has been forwarded to the developers. Let’s see what comes out of that. Of course I replied and thanked them in advance. I also hinted that they might … Continue reading
Oh boy, this thing can’t even count correctly… or since when is 7 = 9 ? Pulse-Trigger: I call that a bug.
This is the first post regarding the Lecroy WaveAce 224 DSO. As it currently seems I have material for at least a few short posts. Currently I’m having issues with one important knob and the triggering system.