Tags64-pixels adapter AVR bulb customs DSO failure by design fireworks firmware good old days hack happy new year hardware IKEA import lamp lecroy LED light lighting linux MBI5168 modding open source opensuse oscilloscope PCB power supply prototype PWM rant remote RGB RoHS Samtid Seeedstudio SMD soldering teardown trigger upgrade WaveAce 224 weller WS2812B youyue
Tag Archives: DSO
This “beauty” appeared on my desk today. Just testing & playing with it. I used the arb sig-gen of a PicoScope 2204A for generating a unpleasant test-signal. Trying to trigger on it was “interesting” :-)
The sender was the current CEO of LeCroy, Thomas H. Reslewic. I’m glad they have finally taken action. I was offered a full refund. I hope they will also look into all the WaveAce 224 issues I’ve documented so far … Continue reading
This is a ‘close relative’ to the ‘eternal boot screen’. It is also caused by the all-mighty triggering system, which governs the whole oscilloscope – even when it shouldn’t. One of the most annoying things is that this can block … Continue reading
Will it ever end? How many bugs are there still to be discovered… A word of warning: there will be some strong language in the post. If anybody affiliated with Lecroy should read this, please feel offended – not as … Continue reading
Another quirk. I seem to have a talent to upset this device. Showing a signal that isn’t there should not happen though.
Automatic cursors are supposed to show what and where stuff is measured on a waveform when automatic measurements are used. This is pretty useful, if it works as expected. Suffice to say that it doesn’t always do that – unfortunately.
Got the ‘pulse-trigger’ to trigger. This time I don’t care if the set pulse-length is actually accurate, I just wanted it to trigger on the ‘thing’ and it did. Fine – happy happy joy joy! Then I felt like moving … Continue reading
I managed to damage the alligator clip on one of my probe’s ground cables. It was probably a bad one and it fatigued. I replaced the alligator clip with one I found in a box. During this procedure it turned … Continue reading
Where art thou? This time I have discovered issues that are trigger-related, but more suitably filed under “general displaying issues”. Still using the “1.01.01.18R2” firmware. Just have a look at the video. No more words…
Oh boy, this thing can’t even count correctly… or since when is 7 = 9 ? Pulse-Trigger: I call that a bug.
To see what of the trigger system of the WaveAce 224 works and doesn’t work (as I expect it), I’ve made a little device, which shall from now one be called DSO-T.
This is the first post regarding the Lecroy WaveAce 224 DSO. As it currently seems I have material for at least a few short posts. Currently I’m having issues with one important knob and the triggering system.
I was having problems with the “led ring” boards. Their internal oscillators were not ticking with the same speed and required constant re-syncing. I tried to improve the situation by recalibrating by adjusting the OSCCAL registers. Recalibration can be done … Continue reading
Maybe you remember my last post about the Jyetech oscilloscope and it’s weird behaviour. Not all of them are strange, as the free replacement DIY KIT I received from them works fine. At that time I suspected a sick ATmega64 … Continue reading
The symptoms since a firmware upgrade (and downgrades, re-upgrades…): strange characters on the display for SEC/DIV system freezes +/- buttons are ignored (but still work physically) the ADC chip doesn’t get a CLK signal during a freeze My theory: The … Continue reading
I bought the DIY KIT at seeedstudio recently, as I needed to feed my soldering obsession. Here are some images of the KIT coming right out of the box: I had to use my milling machine, because the front panel … Continue reading